Create Value for Customer, Make Contribution for World’s Security.
|Place of Origin:||China Beijing|
|Certification:||CE / GOST|
|Minimum Order Quantity:||1 piece|
|Packaging Details:||Standard export carton|
|Delivery Time:||Within 2-3 days after received payment|
|Payment Terms:||T/T, Western Union, Paypal|
|Supply Ability:||500 Unit/Units per Month|
|Relative Humidity:||≤90％||Accuracy:||±(3%H+1) μm|
|Storage:||Memory For Up To 20 Files (up To 50 Values For Each File) Of Stored Values.||Resolution:||0.1μm|
|Measuring Range:||(0～1250) μm, Depends On Probes. Maximum 10mm For The Probe F10.||Display:||4 Digital LCD With EL Backlight|
|Dimensions:||125mm×67mm×31 Mm||Net Weight:||340g|
Portable cars painting Digital Coating Thickness Gauge 4 LCD With EL Backlight
|Measuring principle||Magnetic induction & Eddy current|
|Measuring range||(0～1250)μm, depends on probes, maximum 10mm for the probe F10. Refer to appendix|
|Low range resolution||0.1μm|
|Accuracy||±(3%Thickness+1)μm, depends on probes and conditions|
|Display||4 digits LCD with EL back-light|
|Memory||Memory for up to 20 files(up to 50 values for each file)of stored values|
|Power source||Two “AA” size, 1.5 Volt alkaline batteries, 200 hours typical operating time(EL back-light off)|
The gauge adopts two thickness-measuring methods: magnetic induction method and eddy current method.
Magnetic Induction Method:
The probe and the magnetic metal substrate will form a closed magnetic circuit when probe contacting with the coating, the magnetic resistance of closed magnetic circuit varies due to the existing of non-magnetic coating. The thickness of the coating can be measured through the variation of magnetic resistance.
Eddy Current Method:
The high frequency alternating current generates an electromagnetic field in the probe coil, eddy current will be formed on metal substrate when the probe contacting with the coating, and the eddy current has an effect of feedback on the coil in probe. The thickness of the coating can be calculated through measuring the effect of feedback.
The range of applications is indicated by the probes available.
-F probes work on the magnetic induction principle and should be used for non-magnetic coatings such as aluminum, chrome, copper, zinc, paint and varnish, enamel, rubber etc., on an iron or steel substrate; they are also suitable for alloyed and hardened magnetic steel.
-N probes work on the eddy-current principle and should be used for insulating coatings on all non-ferrous metals and on Austenitic stainless steels, e.g, paint, anodizing coatings, ceramics, etc, applied on aluminum, copper, zinc die-casting, brass, etc.
|Standard Config||1||Main unit||1|
|2||Probe||1||F1 or N1|
|3||Calibration Foils||5 PCS|
|4||Zero plate||1||Iron or Aluminum|
|7||Alkaline Battery||2 PCS||AA size|
|Optional Config||8||Other type of probes||F400, F10, F1(90°),N1|
|9||Data-Pro software||1||On CD|
① The main body
② Key pad
④ Battery cover
⑤ Probe socket
⑥ USB port